Posts Tagged zero calibration
People who have used Milltronics or Siemens ultrasonic continuous level measurement systems are used to the way configuration is done. It’s not uncommon that when they start up their first Siemens radar transmitter they encounter a stumbling block – the 4-20mA output doesn’t respond correctly.
The radar will start up and run, and if the tank level is low, they’ll get a valid level reading on the transmitter’s display, but the 4-20mA will be incorrect (too high a percentage). Or, if the tank level is at a medium or high level, both the indicated level in the display and the output will be maxed out at 20mA. It’s a misunderstanding about where the span measurement starts from. Read the rest of this entry »
20 mA, 4-20 mA, 4mA, calibrating echomax sensor, calibrating Siemens radar, calibrating Siemens ultrasonics, commissioning, Configuration, lower range value, LRV, radar level, span, span calibration, ultrasonic level, upper range value, URV, zero calibration
- Join 482 other subscribers
Search by Topic