Posts Tagged siemens products
A few weeks ago, I wrote about problems a customer encountered trying to move a functioning radar level device from one tank to another. With a little light reading of the user manual, and one change on the radar, the situation was resolved.
People are usually familiar with the concept of false echo suppression: looking at a tank and mapping out the obstructions that can cause false readings, like baffles, ladders, tank braces. But the Siemens products we work with do it a little differently. It’s a built-in feature on Siemens ultrasonic and radar level devices, and only takes five minutes to run.