Posts Tagged radar transmitter
LR560 ignores dust, reports true level
Posted by danstips in Level, Level Technology, Radar, Siemens on January 8, 2016
In a coal-fired power plant, the coal is transported from the yard to the silos on a conveyor system.
The process of dumping coal from the conveyed buckets into the silo creates a lot of dust.
Solving installation problems for radar tank level transmitters
Posted by mikedelacluyse135 in Configuration, Level, Level Technology, Level Transmitters, Radar, Siemens, Transmitters, Troubleshooting on August 25, 2014
Recently, we’ve run into a few radar level applications that had some startup challenges, mostly because the person installing the transmitter didn’t consider how radar wave transmission would affect the level transmitter’s performance.
One trick to any installation is to reduce the number of obstructions encountered by the transmitter. But you have to take into consideration that radar waves don’t transmit in a concentric circle around the transmitter. And this can create a problem of its own.
Why doesn’t my LR560 radar’s analog output match the displayed level value?
Posted by danstips in Configuration, Installation, Level, Level Technology, Level Transmitters, Measurement, Radar, Siemens, Transmitters, Troubleshooting, Uncategorized on September 20, 2012
A customer who had had lots of experience with Milltronics and Siemens ultrasonics was installing his first SITRANS LR560 radar level transmitter. They had worked with it in the shop beforehand, going through most of the settings. They even tested it by setting it up to shoot against a file cabinet and used a tape measure the check the indicated distance value.
Everything checked out OK in the shop.
When they installed the transmitter on the top of the bin, they changed the transmitter’s sensor mode parameter from the distance mode they used in the shop for testing to level mode. After aiming, the level value shown in the local display was dead nuts on, but the 4-20mA signal going back to the control room was way off.
The bin was a third full. The 4-20mA showed it about double that. Not only that, the 4-20 was going in the wrong direction. The bin was emptying and the HMI reported an increasing level value. Someone realized that an inverse-acting output was typical of a distance value, so they reconfigured the sensor mode to distance. That got the 4-20mA much closer to a distance value, but it was still not exactly what it should be, and besides, the goal was to read level, not distance, in the control room. What was going on?
February 2: A day to look back
Posted by danstips in Air Bubbler, Communications, Configuration, HART, Honeywell, Installation, Level, Level Technology, Paperless Recorders, Pressure, Pressure Switches, pressure switches, Pressure Transmitters, Siemens, Software, Switches, Transmitters, Trendview X-Series Paperless Recorders, Troubleshooting, United Electric on February 2, 2012
Yesterday and today, people all over Chicago and the Midwest were looking at pictures from last year. We had 22″ of snowfall in one day. The roads were closed. The airports were closed. Even the Lesman offices were closed. And today’s weather forecast? 45°… in Chicago… in February.
This morning in Pennsylvania, a groundhog named Phil came out, looked back, and saw… his shadow.
All this looking back made me a little reflective myself. I’ve been writing this blog for about 6 months now. So I thought I’d take a minute and recap the articles people keep coming back to read: