Posts Tagged process instrumentation
Chemical seals and fills are often necessary to protect your process instrumentation from harm. But the wait for a custom gauge-and-seal, switch-and-seal, or transmitter-and-seal combination could be an issue, keeping your process offline for longer than necessary, or costing a premium for quick delivery.
We’ve removed the potential for downtime and rush charges with our WIKA-certified assembly station, and improved stock of most popular transmitters, switches, gauges, diaphragm seals, and fill fluids. Lesman customers can experience next-day delivery on custom assemblies from in-stock instrumentation.
People have asked me about setting fault alarms in level transmitter analog signals at 2mA or 3mA levels. What they typically don’t understand is that a two-wire transmitter uses the electrical current below 3.6 mA for its own power and operation. So, a 2.0 mA or 3.0 mA fault indication just isn’t possible. At these low currents, there wouldn’t be enough power to generate the fault indication signal and to keep the transmitter functioning properly. Read the rest of this entry »