Posts Tagged pressure transmitter
Tips for ranging a vacuum transmitter
Posted by danstips in 4-20mA, Honeywell, Pressure, Pressure Gauges, Pressure Transmitters, pressure transmitters, Siemens, Transmitters, WIKA on July 31, 2014
A process plant’s technician was mystified about how to get a typical gauge pressure transmitter to read in the vacuum range. “All our gauges are 0 to 30 inches mercury, and that’s what we need to transmitter output to be. But the transmitter you sent us just stays around 4mA when we pull a vacuum.”
We walked out to the reactor vessel to look at the installation. The transmitter’s Low side port was open, its high side port was plumbed into a tee along with a conventional bourdon tube pressure gauge reading gauge pressure vacuum.
I could see why he was confused. The mechanical gauge goes from 0 to 30. I asked what range he used to configure the pressure transmitter. His answer, “0 to 30 inches mercury, same as the mechanical gauge.”
So, what was happening?
Honeywell smart transmitter design makes communication card swap easy
Posted by danstips in Communications, Configuration, HART, Honeywell, Honeywell DE, Pressure Transmitters, pressure transmitters, Service, Transmitters, Troubleshooting on January 28, 2014
It was bound to happen sooner or later.
I took a call from a customer who needed to replace a garden variety differential pressure transmitter… with one exception: He needed Honeywell’s DE digital protocol for communicating to his DCS. The DE protocol is still great, but since so many installations today use HART or Foundation Fieldbus, all of our in-stock pressure transmitters had a HART communication card – a critical mismatch to what the customer needed.
A year ago, we would have been stuck rush-ordering a unit from the factory, with all the attendant delays and expediting charges, because you couldn’t swap out a comms card without making the transmitter’s hazardous approval invalid.
What could we do?
Read the rest of this entry »
Wet rhymes with Set (or how to remember LRV/URV configuration prompts)
Posted by danstips in Configuration, Honeywell, Installation, Level Transmitters, Pressure, Pressure Transmitters, pressure transmitters, Siemens, Transmitters on July 26, 2013
Lots of people like the pushbuttons on industrial pressure transmitters because the basic settings that every transmitter needs can be set up without a HART communicator. This includes things like the tag name, engineering units, LRV (Lower Range Value, the zero, or what 4.0mA represents), URV (Upper Range Value, the span, or what 20.0mA represents) and damping (an average or filter factor that dampens noise).
On the new Honeywell ST700/ST800 series smart transmitters, the tag name and engineering units are easy to configure and self explanatory, but I seem to stumble when setting up the LRV and URV because I’m faced with a non-descript choice. There’s two sets of options (under Transmitter Setup, not Calibration):
OK, either configures an LRV or a URV value, but which is which? What’s the difference?
How can I keep the water out of my transmitters in washdown conditions?
Posted by danstips in Installation, Measurement, Pressure, Pressure Transmitters, Siemens, Transmitters on July 12, 2012
Earlier this week, I took a call from a customer having problems with his pressure transmitter. He’d figured out that his last transmitter, installed in an area with nearly continuous wash-downs, had failed because of water intrusion. The bigger problem was that it wasn’t coming from a loose cover, a conduit pipe, or a leaky conduit connection. In his case, the water came into the transmitter through a vent hole and messed up the measurement cell electronics.
So he asked me to recommend a 3000 PSI transmitter that could survive the washdown conditions.