Posts Tagged pressure pulse
A while ago, I got a call from a customer who was having trouble with a differential pressure transmitter. He was using a draft range DP transmitter to measure the pressure in a combustion chamber, so it could be controlled with a damper. He had one port connected to the combustion chamber with impulse tubing, and the other (low side) was left open to the atmosphere.
He’d noticed that when a fork truck or other vehicle sped past the furnace – the transmitter was mounted next to a traffic lane — it cause the furnace pressure to momentarily dip downward, and cause the damper to oscillate.
He figured out that the air movement provided by the passing vehicle was creating a momentary pressure pulse on the low side port. These air movements were creating difficulty in maintaining furnace pressure.
So, he asked me, “How can we dampen the effect of the momentary pressure pulse?”