Posts Tagged dp transmitter
Minimizing shift on a draft range differential pressure transmitter
Posted by danstips in Honeywell, Pressure Transmitters, pressure transmitters, Transmitters, Troubleshooting on August 19, 2014
A while ago, I got a call from a customer who was having trouble with a differential pressure transmitter. He was using a draft range DP transmitter to measure the pressure in a combustion chamber, so it could be controlled with a damper. He had one port connected to the combustion chamber with impulse tubing, and the other (low side) was left open to the atmosphere.
He’d noticed that when a fork truck or other vehicle sped past the furnace – the transmitter was mounted next to a traffic lane — it cause the furnace pressure to momentarily dip downward, and cause the damper to oscillate.
He figured out that the air movement provided by the passing vehicle was creating a momentary pressure pulse on the low side port. These air movements were creating difficulty in maintaining furnace pressure.
So, he asked me, “How can we dampen the effect of the momentary pressure pulse?”
How do I set upper range value and lower range value on a Siemens pressure transmitter?
Posted by danstips in Configuration, Installation, Level Technology, Pressure, Pressure Transmitters, Siemens, Transmitters on February 21, 2012
A customer needed to re-range a Siemens SITRANS P DS III transmitter without applying pressure. He’d read my earlier blog post about using the pushbuttons to program the DS III (Pressure Transmitter Pushbuttons to the Rescue!), and where to find the Z-fold instruction sheet.
All was going to plan, until he started looking for the settings for lower range value (LRV) and upper range value (URV). And then, in an all-too-familiar scenario, he hit a roadblock. The instructions don’t actually mention URV and LRV, the terms he’s familiar with, by name or acronym. So what’s a guy to do?