Archive for category Flowmeters
How to install magnetic flowmeters in less than optimum conditions
Posted by danstips in Configuration, Flowmeters, Installation, Magnetic Flowmeters, Siemens, Webinar on October 16, 2015
Not all processes have optimal piping configurations. Learn how to identify alternative solutions for processes inhibited by piping constraints.
Siemens is hosting a free webinar on Wednesday, October 21 at 2pm EST.
In this 1-hour webinar, Siemens specialists will show you how the flowmeter can be installed and how this versatile instrument can be used to meet your application needs. They will be discussing the installation guidelines for process piping that can enhance your flowmeters’ performance.
Making a Siemens clamp-on flowmeter setup record easy to read
Posted by danstips in Communications, Configuration, Flow, Flowmeters, Serial, Siemens, Software, Troubleshooting, Ultrasonic Flowmeters on October 1, 2014
Everyone agrees that it’s good practice to keep a record of configurations and setups for any field instrument. I’m constantly recommending it to our customers for their transmitters, controllers, recorders, and other complex configurable devices.
Siemens clamp-on ultrasonic flowmeters even have a system in place to make this process easy. By connecting the meter to a PC through the RS-232 serial port, you can use a terminal program and the SITE command to fetch a data file that holds all the instrument’s configuration data.
The terminal program can also be used — with a set of instructions specific to the flowmeter — for viewing real-time operational data, performing device setup, uploading logger data, or uploading configurations known as SITE setups.
Recently, I was called to visit a plant and look at a misbehaving flowmeter. From previous discussions with the operator, I knew he’d saved SITE setup files for every flowmeter installed in the plant.
I asked if the customer would e-mail me the setup file before my visit, so I could check out how the flowmeter was set up. My request was met with a chuckle and “Well, if you really think it’s worth it…”
A “So What” review of Siemens new SITRANS LUT400 ultrasonic level controller
Posted by danstips in Control, Echomax transducers, Flowmeters, Level, Level Technology, Open Channel Flow, Siemens, SO WHAT Reviews, Ultrasonic on November 5, 2012
Part of my job as the technical specialist at Lesman is to make sense of new products and upgrades, and figure out what’s really going to matter most to our customers.
Sometimes it’s just a matter of a much-needed new functionality, or better configuration tools, easier mounting, or switching to the most current form of data storage.
In the case of Siemens’ latest ultrasonic controller, it’s all that and more.
Siemens (and Milltronics) ultrasonic controllers and transceivers, like the HydroRanger, MultiRanger, and OCM-III have been around for years with no significant improvements. Instead of updating these devices, Siemens has done a complete redesign, and introduces the SITRANS LUT400 as the first device in the new ultrasonic controller family.
Here are my initial thoughts on this new player in the ultrasonic game.
Why is my clamp-on flowmeter’s flow rate stuck at zero?
Posted by danstips in Configuration, Flow, Flowmeters, Repairs, Service, Siemens, Ultrasonic Flowmeters on August 28, 2012
I was working with a customer to replace a Siemens clamp-on ultrasonic flowmeter after its electronics had been replaced. We followed all the connection and startup instructions, but the flow rate was stuck at 0.00 – no dithering, no hunting, not a single flicker in the low-order digits.
Ultrasonic clamp-on flowmeters can be very sensitive to low flows. But even at no flow with the pump turned off or the line isolated, the digits to the right of the decimal point would dither around zero, showing some very small positive values, some very small negative values. It’s the nature of the beast.
So why were we getting an absolute zero reading? What was going on?