Archive for category Honeywell
Last week, I was doing a firmware update on the Honeywell SLG700 guided wave radar using Honeywell’s Smart Anytime Tool (SAT).
Honeywell’s SAT firmware update tool needs to be “pointed” to the component type (sensor/display/communications module), and then the specific file needed for update. A recent release (R102.1) for the SLG700 listed two different files for the Advanced Display, but it wasn’t clear which file I needed to use. Read the rest of this entry »
People will fire up a new UDC 1/4 DIN loop controller, like a UDC2500 or UDC3200, and discover that the lower display periodically flashes an error code: UNPLUG. But what does it mean? Is something wrong?
A search of the pdf version of the manual hints that the Modbus Ethernet communications is involved:
The Ethernet status screen shows the network status of the Ethernet Link. This may be accessed either via Ethernet or via Infrared Communications. For example, if the Ethernet cable is unplugged, then the instrument cannot send up the EUNPLGED diagnostic message via Ethernet.
But the word UNPLUG is nowhere else in the manual.
Written by: Dan Weise
It’s been about a decade since the Honeywell Trendview X-Series paperless recorders were first introduced. And as with any product, you can expect some routine maintenance and service requirements that come with age.
We’ve summarized the top four obstacles you may face when tending to your aging paperless recorder, and how you can overcome them to further maximize the lifespan and functionality of your device.
Before you start, make sure you have a printed record of your paperless recorder’s custom configurations in case you need to restore them to your unit.
1. Changing the Clock Battery
If your paperless recorder is getting old, the battery may be ready to be swapped out. Your recorder uses a standard CR2032 lithium battery that is both inexpensive and readily available at most drug stores.
Although this appears to be a simple solution, I’ve noticed when I change out the battery, the system resets the clock to January 1, 2003.
If you haven’t enabled security, you can just login and change the time. But, this can pose a problem if your recorder uses the ESS extended security (usually a pharmaceutical requirement). When an ESS-enabled recorder resets back to Jan 1, 2003, no user, not even an administrator, can get access to the setup because the valid period for all passwords appears to have expired.
In order to avoid this problem, it is best to plan ahead before attempting to replace the clock battery.
Honeywell’s Process Instrument Explorer (PIE) configuration software used for configuring UDC controllers and the UDA analytical controller communicates with the instruments via RS-485, Ethernet, or Infrared (IR).
Since most newer PCs don’t ship with built-in serial ports to connect an IR adapter, you can use a USB-to-RS-232 converter, and then connect using the Actisys serial-to-infrared adapter (ACT-IR220L+). The USB converter will plug directly into your PC’s USB port, but install on a virtual COM port.
Here’s where it gets tricky: That COM port has to match the COM port used in PIE, and PIE doesn’t support ports above COM8.