Archive for category batch control
A customer has several UDC 3200 loop controllers with newly added Ethernet cards. He needed to configure each of the controllers’ IP addresses using Honeywell Process Instrument Explorer (PIE) software. Because the controllers are working in a 24×7 continuous process, he was concerned about how making those changes would affect each controller’s performance.
So he asked me: Would a PIE action of uploading config files from or downloading them back to a controller affect the controller’s performance?
In the past, I’d only ever changed a controller’s IP address when it was on my workbench, not when it was actively controlling a process. So I’d never paid attention to whether PIE communications would affect the controller’s output or its PID action. Since I couldn’t answer the question, I told the customer I’d run a test to find out for sure.
If you use thermocouples in high-temperature applications, you’re aware of the issues thermocouple drift can cause. Thermocouples drift. It’s not a question of IF, it’s a question of WHEN. And thermocouple drift costs processors time and money in processing errors, waste, downtime, and lost production.
Thermocouple drift occurs due to metallurgical changes of the metal alloy elements over the extended use of the sensor. Thermocouples can drift by as much as several degrees per year.
Earlier this week, I was working with one of our Honeywell Trendview X-series paperless recorders. The X-Series is capable of accepting data input from a USB barcode scanner — typically used for batch identification or for user name logins — but mine wasn’t working right.
When I tried to scan a barcode, I got an error message, and it stumped me. Luckily, I figured it out in a few minutes.
Industry surveys say that nearly half of all processes aren’t accurately tuned. If you read my post on accuracy, stability, and repeatability, you’ll know that a poorly tuned process can result in bad readings, downtime, and wasted materials.
If you use a Honeywell UDC2500, UDC3200, or UDC3500 1/4 DIN universal digital controller, there’s a great built-in function called Accutune that can help make sure your control process is properly tuned.